Navigate to Setup > Equipment > New.
Input a Configuration name.
Select Range pole from the Machine type drop-down list.
Select Prism from the Sensor drop-down list.
Select Top of pole from the Mounting location drop-down list.
Select the Units that you are working in.
Select GT from the Instrument drop-down list.
Input the Vert. prism height .
Input the Prism constant.
Select the Prism type.
Select Bluetooth (Windows BT) as the Connection.
Select RC5/Direct to TS (STD mode) from the Connect via drop-down list.
- Note: If using LPS, you need to select RC5/Direct (Ext. Link).
Check the Use RC box.
Select 9600 from the Baud Rate drop-down list.
Select 8 from the Data Bits drop-down list.
Select 1 from the Stop Bits drop-down list.
Select None from the Parity drop-down list.
You will see a message that machine configuration is complete.
Verify the selected machine, and click Ok.
Navigate to Setup > Station Setup.
Choose from the Setup method drop-down list.
Enter the Instrument height.
Select from the Units drop-down list.
In the Bluetooth Devices window, select RC-5A.
Pocket 3D will connect to the RC-5A and then to the total station through the RC-5.